A: Bubble (or level vial) sensitivity is usually defined by manufacturers as an angular change in the position of the axis of the bubble per 2 mm of bubble movement. On most tubular (plate) vials, graduations are marked at 2 mm intervals. On circular vials, it is common to have an inscribed circle that has a 2 mm radius (4 mm diameter). When you say that some poles are “easier” to hold than others, it is because the easier poles have less sensitive circular vials. While not all-inclusive, the sensitivity of prism/antenna pole bubbles tend to be in the range of 30 to 60 arc-minutes. Thus, easier-to-hold poles may have circular vials in the 60' range and the harder ones may be in the 30' range. When a pole appears to be easier to hold, it simply means that you see less movement of the bubble itself for a particular amount of movement of the prism or antenna than with a more sensitive vial. The sensitivity of the vial (whether circular or tubular) is controlled by the curvature of the glass; the larger the radius, the more sensitive the bubble.
It is a good idea to evaluate the sensitivity of your level vials if you aren’t able to get information from the manufacturer, so that you can determine the impact of the position uncertainty of the prism or antenna in your survey. It can also help you determine whether you need to take extra measures such as using a brace pole to help steady the prism/antenna pole, or a tripod or bipod.
Determining uncertainty of the position of the prism or antenna is quite simple once you know the level vial sensitivity and your observation of the average “out of center” position of the bubble during the typical observation. Figure 1 shows a circular vial with the bubble centered in the inscribed circle. If your typical average is as represented by the bubble position in Figure 2, then the center of the bubble is 2 mm out of position with the center. (Figure 3 shows a tubular vial with the bubble 2 mm out of center.) The distance error would be determined by multiplying the tangent of the angle error represented by the bubble position by the height of the prism pole. If the vial has 60' sensitivity and the pole is 6 ft tall, the horizontal mis-centering would be 0.105 ft (tan 60' x 6 ft). The direction of the error would be random and, in the case of total station measurement to a prism, would create a maximum distance error when the bubble is out of center along a direction that is parallel to the line being measured. If the bubble is out of center at right angles to that direction, the maximum angular error would result. In actual practice it would be a combination of the two.
GIAA is a trade association of manufacturers, suppliers and distribution partners encompassing the present and emerging technologies in surveying, GPS, engineering, construction, GIS/LIS and related fields.
Leica Geosystems Inc.
Trimble Navigation Limited
The American Surveyor magazine
Professional Surveyor magazine
Crain Enterprises Inc.